Aeolos valved holding chamber infant 0-18 months (Mask + Mouthpiece)
Aeolos is an anti-static holding chamber with one way valve and it is used for effective drug delivery with MDI (Metered dose inhaler).By using aeolos is achieved much less quantity of medicine in the mouth and throat and even more medicine directly into the lungs where it is needed.Aeolos due to its small shape and the inhalation control technology that it is using ,is your best ally in asthma treatment.
- 2 products in 1 : Mask & mouthpiece in one package.
- BPA FREE (non toxic plastic).No bisphenol-A.
- Easy to use due to its small shape.
- Big chamber capacity 175ml.
- Specially designed low resistance valve (butterfly type) makes drug delivery easier.
- Exhalation valve on the mask.
- Signal whistle at the backpiece alerts when breathing very fast/when drug delivery is ineffective.
- MDI adapter at the backpiece that fits all prescribed MDIs.
- Body made of transparent material allows visual contact inside the chamber.
- Available in three sizes of mask:infant (0-18 months),child (1-6 years),adult (6+ years)
Instructions For Use AEOLOS
1. Before use, the device must be checked inside and if any defect is detected the device must be replaced.
2. Remove the cap from the MDI (Metered Dose Inhaler), shake well and insert it at the backpiece of the chamber.
3. Apply the mask over the face and make sure that it will be sealed during inhalation.
4. Press the MDI and make sure that the patient is inhaling slowly.If you hear the flow signal whistle sound then the inhalation must slow down.During normal inhalation, the inhalation indicator must move toward the patient.
5. The mask must be applied over the patient's face for 5-6 breaths after providing one puff.
Cleaning Instructions AEOLOS
1. Remove the backpiece of the chamber.
2. Soak all the parts of the chamber in lukewarm water with mild detergent solution. WARNING! Avoid to scrub the parts of the chamber with any kind of cloth or sponge.
3. Rinse all the parts in clean water. Or remove all the parts of the chamber and place them in the top rack of the dishwasher (up to 70ο C) in a vertical position.
4. Place all the parts of the chamber in vertical position and let air dry. WARNING! Avoid rub drying.
5. Before reassemble the chamber make sure that all the parts are dry.
* The spacer should be replaced after one year use.